Instrumentation and Detectors (physics.ins-det)

• The prospect of pileup induced backgrounds at the High Luminosity LHC (HL-LHC) has stimulated intense interest in developing technologies for charged particle detection with accurate timing at high rates. The required accuracy follows directly from the nominal interaction distribution within a bunch crossing ($\sigma_z\sim5$ cm, $\sigma_t\sim170$ ps). A time resolution of the order of 20-30 ps would lead to significant reduction of these backgrounds. With this goal, we present a new detection concept called PICOSEC, which is based on a "two-stage" Micromegas detector coupled to a Cherenkov radiator and equipped with a photocathode. First results obtained with this new detector yield a time resolution of 24 ps for 150 GeV muons, and 76 ps for single photoelectrons.
• Fast digitisers and digital pulse processing have been widely used for spectral application and pulse shape discrimination (PSD) owing to their advantages in terms of compactness, higher trigger rates, offline analysis, etc. Meanwhile, the noise of readout electronics is usually trivial for organic, plastic, or liquid scintillator with PSD ability because of their poor intrinsic energy resolution. However, LaBr3(Ce) has been widely used for its excellent energy resolution and has been proven to have PSD ability for alpha/gamma particles. Therefore, designing a digital acquisition system for such scintillators as LaBr3(Ce) with both optimal energy resolution and promising PSD ability is worthwhile. Several experimental research studies about the choice of digitiser properties for liquid scintillators have already been conducted in terms of the sampling rate and vertical resolution. Quantitative analysis on the influence of waveform digitisers, that is, fast amplifier (optional), sampling rates, and vertical resolution, on both applications is still lacking. The present paper provides quantitative analysis of these factors and, hence, general rules about the optimal design of digitisers for both energy resolution and PSD application according to the noise analysis of time-variant gated charge integration.
• The MAJORANA Collaboration is searching for the neutrinoless double-beta decay of the nucleus Ge-76. The MAJORANA DEMONSTRATOR is an array of germanium detectors deployed with the aim of implementing background reduction techniques suitable for a tonne scale Ge-76-based search (the LEGEND collaboration). In the DEMONSTRATOR, germanium detectors operate in an ultra-pure vacuum cryostat at 80 K. One special challenge of an ultra-pure environment is to develop reliable cables, connectors, and electronics that do not significantly contribute to the radioactive background of the experiment. This paper highlights the experimental requirements and how these requirements were met for the MAJORANA DEMONSTRATOR, including plans to upgrade the wiring for higher reliability in the summer of 2018. Also described are requirements for LEGEND R&D efforts underway to meet these additional requirements.